Getting the kind of accuracy you want isn't easy, but it can be done.
Dear Control Engineering: I saw a photo of a flowmeter testing setup that seemed hopelessly complicated. What’s the point of that, and is it really necessary?
The photo you saw may have been the new ABB testing facility in Shanghai, China. ABB has a large instrumentation manufacturing facility there, and that normally requires appropriate testing capabilities. In this particular case, they have built in support to test everything from ½ to 96 in. diameter flowmeters, so that means they need the ability to control flow precisely from 0 to 50,000 gpm.
Testing flowmeters is a difficult process, because there is no practical way to do it without simply pushing whatever volume of liquid you need through the device. To make matters worse, you have to calibrate one flowmeter with another one that is even more accurate. If you follow the old rule of thumb, you need a measuring device that is capable of ten-times the accuracy of the unit you want to check. In other words, if you have to calibrate a flowmeter to ±1 gpm, the flowmeter that is your standard has to be able to read ±0.1 gpm. Given the high accuracy of instrumentation these days, that can be a serious challenge. I have seen flowmeters that are used as calibrators that are capable of accuracy of ±0.015%. Once you get to this level, calibration involves measuring liquid volumes and other ways to boost the practical accuracy of a device.
But that’s just the start. The dynamics of liquid flowing through a pipe are messy with turbulence, velocity profiles, entrained gas, and the like. If you study that picture, you’ll see long sections of straight, horizontal pipe that minimize those effects. Each pipe joint will be smooth to avoid any obstacles that will disrupt laminar flow. I’ll even wager that the water itself is calibrated, in that its chemical composition is analyzed, its precise density known, and temperature regulated. There will also be critical weighing apparatus to measure specific volumes of water. All of these are there to eliminate any variability to the process.
The new lab in Shanghai is approved by the Shanghai Institute of Measurement and Test Technology (SIMT), which is itself accepted by a group of international testing bodies, including UKAS, NIST, DKD, NVLAP, NATA, and ILAC. All things considered, it is a very complex process but necessary to deliver the kind of accuracy that users demand.
Peter Welander, pwelander(at)cfemedia.com
|Search the online Automation Integrator Guide|
Case Study Database
Get more exposure for your case study by uploading it to the Control Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.