Mechanical flowmeter family for water service
Omega introduces FTB8000HW Series, able to tolerate low-quality and hot water.
Omega has introduced its FTB8000HW series of hot-water meters. This series features flow rates from .22 to 132 gpm, with 150 psi maximum operating pressure and 90 °C (194 °F) maximum operating temperature. Units in the series use the multi-jet principle, which has been a widely accepted standard for many years. This type of meter mechanism is known for its wide range, simplicity, and accuracy in low-quality water. The impeller is centered in a ring of jets, with inlet on one level and outlet on another. A gear train drives the register totalizer dials. For pulse output, one of the dials is replaced by a gear, which turns a magnet that is detected by an encapsulated sensor threaded into the outside of the lens. Pulse rate is determined by the gear and the dial on which the gear is placed. Changing the pulse rate requires no special tools and can be done in the field.
Mechanically, all FTB8000HW meters are the same. The difference between the -PR, -PT, and standard models is in the sensor. -PR model meters use a solid state, Hall-effect sensor, which requires power. -PT model meters use a 2-wire reed switch which provides a dry-contact closure and does not require power. Basic meters totalize only and do not have a rate output.
Edited by Peter Welander, pwelander(at)cfemedia.com
Case Study Database
Get more exposure for your case study by uploading it to the Control Engineering case study database, where end-users can identify relevant solutions and explore what the experts are doing to effectively implement a variety of technology and productivity related projects.
These case studies provide examples of how knowledgeable solution providers have used technology, processes and people to create effective and successful implementations in real-world situations. Case studies can be completed by filling out a simple online form where you can outline the project title, abstract, and full story in 1500 words or less; upload photos, videos and a logo.
Click here to visit the Case Study Database and upload your case study.