# Pressure gage ranging and accuracy

08/29/2010

Dear Control Engineering: I was reading the recent article on differential pressure sensors, and I had a question coming out of that discussion. The author made the following statement: “The pressure in the line may be upwards of 1,000 psi, but the pressure drop is usually very small. Using two separate gage pressure devices and subtracting the readings would result in the combined error of two devices having wide measuring spans. The result would likely be a very poor measurement seldom suitable for control.” Could you explain that in greater detail?

That statement was made in the context of using a DP (differential pressure) sensor to measure flow using an orifice plate flowmeter. His point is that a single DP device is better than trying to do the same job with two pressure gages and calculating the difference.

Here’s why: Using his example, let’s say your process line is running with a pressure of 1,000 psi, and you have an orifice plate. Under normal operating conditions, the pressure drop across the plate is 5 psi. That means if everything is perfect, the upstream gage reads 1,000 psi and the downstream gage reads 995 psi. For appropriate ranging, both gages are identical with a scale of 0-1,500 psi. The problem emerges when you consider the accuracy of the two devices. If they have an accuracy rating of ±1% full scale, that means any reading can be high or low by 1% of the full scale, or 15 psi. So the first gage could indicate 1,000 psi as anything from 985 to 1,015 and still be in tolerance. Similarly, the second gage could indicate 995 psi as anything from 980 to 1,010 and still be in tolerance. Do you see the problem? You’ll never be able to know when your pressure drop is, in fact, 5 psi.

The differential pressure device, if it has a range of 0 to 15 psi, will give you a much higher accuracy since 1% of 15 is 0.15 psi.

Now, the tricky part is making sure you’re gage isn’t exposed to 1,000 psi on one side and 0 on the other. This could damage it, so you will have to ensure that the device won’t see pressure level imbalances outside of its ability to withstand them.

Peter Welander, pwelander@cfemedia.com

Also read an earlier article on pressure measurements. article

The Engineers' Choice Awards highlight some of the best new control, instrumentation and automation products as chosen by Control Engineering subscribers. Vote now (if qualified)!
The System Integrator Giants program lists the top 100 system integrators among companies listed in CFE Media's Global System Integrator Database.
Each year, a panel of Control Engineering and Plant Engineering editors and industry expert judges select the System Integrator of the Year Award winners in three categories.
This eGuide illustrates solutions, applications and benefits of machine vision systems.
Learn how to increase device reliability in harsh environments and decrease unplanned system downtime.
This eGuide contains a series of articles and videos that considers theoretical and practical; immediate needs and a look into the future.
HMI effectiveness; Distributed I/O; Engineers' Choice Award finalists; System Integrator advice; Inside Machines
Women in engineering; Engineering Leaders Under 40; PID benefits and drawbacks; Ladder logic; Cloud computing
Robotic integration and cloud connections; SCADA and cybersecurity; Motor efficiency standards; Open- and closed-loop control; Augmented reality
Programmable logic controllers (PLCs) represent the logic (decision) part of the control loop of sense, decide, and actuate. As we know, PLCs aren’t the only option for making decisions in a control loop, but they are likely why you’re here.
This digital report explains how motion control advances and solutions can help with machine control, automated control on assembly lines, integration of robotics and automation, and machine safety.
This article collection contains several articles on how advancements in vision system designs, computing power, algorithms, optics, and communications are making machine vision more cost effective than ever before.
Control room technology innovation; Practical approaches to corrosion protection; Pipeline regulator revises quality programs
Cloud, mobility, and remote operations; SCADA and contextual mobility; Custom UPS empowering a secure pipeline
Infrastructure for natural gas expansion; Artificial lift methods; Disruptive technology and fugitive gas emissions
Automation Engineer; Wood Group
System Integrator; Cross Integrated Systems Group
Jose S. Vasquez, Jr.
Fire & Life Safety Engineer; Technip USA Inc.
This course focuses on climate analysis, appropriateness of cooling system selection, and combining cooling systems.
This course will help identify and reveal electrical hazards and identify the solutions to implementing and maintaining a safe work environment.
This course explains how maintaining power and communication systems through emergency power-generation systems is critical.